Nurse Call System ...
to ensure Patient Care
Nurse Call System - Esteem
Models of Nurse Call System:
Working of NCS-EsteemBy turning ON the switch on the calling unit an indicator gets lighted up confirming that your call is passed on, which in turn reaches the Controller/Display unit and flashes the room number of the patient on it. simultaneously, the Voice module in it announces the call which forces the nurse to give immediate attention.
After noticing the call the nurse can silence the voice module by pressing the MUTE button on Controller/Display Unit which makes the flashing window steady but it will continue glowing unless it is cleared by the nurse from the patient”s room using the calling unit. If another patient raises a call, again the same procedure takes place independent of the preceding call.
Thus NCS ensures a positive and immediate response from the nurse for the patient needs.
This is a Microcontroller based device placed in the nurse station with static windows which displays the room number of the calling patient and also it has a voice module to announce the call. There are three buttons on it,
Power Button: Pressing this for long time(approximately 3 sec) turns the device ON or OFF.
MUTE Button: It is for acknowledging a call from patient. It will make the flashing window steady and also silences the voice module until a call is cleared.
Test Button: On pressing the button it will flash all the windows and will announce ” Attention, Patient calling ” to make sure that the device is working fine.
Electronic Calling Unit
The unit is designed with soft touch key pad consisting of one key for call initiation and another for clearing call which is required to be done by the nurse attending to the patient.
Additional push button switch is integrated to this unit by means of a coil cord to enable a patient to initiate a call from the bed itself.
Toilet Calling Unit
Here call is originated by pulling the cord and can be cleared by pulling it again. The LED indicator lights up if call is active. This allows the patient to initiate a call from squatting positions itself.
These semi circular shaped indicators can be mounted on the wall near the patient's door so that , nurse walking through the corridor can see it from a distance.These are made of super bright LEDs to indicate patient's call pending which can be seen from the corridor itself.
230V Interface Module
The interface unit to convert existing wiring (laid to work with 230V AC Buzzer) to 12V DC operated ” NURSE CALL SYSTEM ”. This interface and Nurse Call System can replace the existing buzzer system instantly without any change in cabling but providing for patient safety which was absent earlier.
Models of Nurse Call System:
Indicators : Flashing 5mm LED to indicate NCS is active.
Power : 230 V AC +/-10%,50Hz, 15 watts
Front Panel Buttons